Nibar When a trigger is set up properly, the oscilloscope converts unstable displays or blank screens into meaningful adscal. Taking Cursor Measurements 3. The readout on the top right corner which unit is sec, show us the horizontal position. Note The oscilloscope can capture a waveform in normal YT mode at any sampling rate.
|Genre:||Health and Food|
|Published (Last):||20 September 2012|
|PDF File Size:||4.85 Mb|
|ePub File Size:||6.19 Mb|
|Price:||Free* [*Free Regsitration Required]|
Nibar When a trigger is set up properly, the oscilloscope converts unstable displays or blank screens into meaningful adscal. Taking Cursor Measurements 3.
The readout on the top right corner which unit is sec, show us the horizontal position. Note The oscilloscope can capture a waveform in normal YT mode at any sampling rate. You cannot adjust the horizontal position and scale of the reference waveform. Contents in this Manual are not allowed to copy, extract and translate before being allowed by Atten Company.
Screen Normal set to normal mode. Automatically sets the oscilloscope controls to produce a usable display of the input signals. This aspect has been sent to the manufacturer and they said they would work on it. Chapter 4 prompting messages and troubleshooting 4.
Mention you that the trigger Level is at a limit when you turn the Trig level knob. Record Press this button to enter the waveform record menu. Turn off the Digital Filter function. Do not touch exposed connections and components when power is present.
The cross cursor sets the position on the waveform automatically. To avoid electric shock when using the probe, keep fingers behind the guard Note on the probe body. Contact the site with comments or questions. Support the sampling rate: The shapes with two arrows beside them represent the front panel knobs.
Page 6 General safety summary Stten Exposed Circuitry. MATH function control button. We should have purchased a Rigol scope instead. Software version List the software version. Chapter 5 Service and Support It is also a arrow on the scale top to show the vertical position.
The oscilloscope has seen a trigger and is acquiring the posttrigger data. Taking Simple Adscap 3. When Select trigger condition. When Select how to compare the Positive pulse width less than pulse trigger pulse relative to the width setting value selected in addcal Set Pulse Width option.
Page 41 If you input a sine signal to channel 1, follow these steps: Page Picture To record waveforms, follow these steps: Save Mahual created mask setting. Same with the software. Peak Detect mode capture the maximum and minimum values of a signal Finds highest and lowest record points over many acquisitions. Because the measurements use the waveform record points, they are more accurate than the graticule or cursor measurements. Most Related.
You suspect that the signal contains much more detail than you can now mxnual in the display. Probe Attenuation Setting 1. This can result in aliasing may cause narrow pulses to be missed. Set up the channel bandwidth limit Take qdscal CH1 for example; the tested signal is a pulse signal with the high frequency surge: Page 56 Picture Operate Instruction: Page Picture Table Mask Setting function menu 1: If you suspect there is damage to this product, have it inspected by qualified service personnel. Pop-up Menu display pattern is more convenient for users using it. It is also a arrow on the scale top adscsl show the vertical position. In this mode, you can observe two irrelative signals at the same time.
Atten ADS1000 Series Manual
This can result in aliasing may cause narrow pulses to be missed. Picture Table Mask Setting function menu 1: In this mode, the waveform display updates from left to right. My Subscriptions Subscriptions Help. Update Firmware Picture 2. Page — Chapter 4 prompting messages and trouble Acquiring Signals System 2.
Digital Oscilloscope ATTEN ADS1102CAL+